Colorectal cancer (faecal occult blood) test
About the parameter
The presence of haemoglobin (the major protein in red blood cells) in faeces is indicative for the presence of blood (faecal occult blood, iFOB) as a result of bleedings associated to pathologies of the gastrointestinal track such as colon polyps, adenomas, colorectal carcinoma, ulcerative colitis and Crohn’s disease.
Colorectal cancer (CRC) is the third most common form of cancer in the western world, however, it often causes no symptoms until it has reached a relatively advanced stage. Thus, many organisations recommend the screening of this disease through iFOB testing.
Precise, immunological assay principle
Quantitative immunological stool screenings using the same principle as the Eurolyser iFOB colorectal cancer test show significantly better values in screening colorectal neoplasiae, compared to a classic haemoccult test.
Colorectal bleeding can sometimes be intermittent, therefore negative results do not necessarily rule out the disease; a colonoscopy is recommended as a definitive test! An iFOB test is also not suitable as screening method for diseases of the upper gastrointestinal tract.
Polyclonal antibody haemoglobin detection in faeces. The latex particles coated with antibodies anti-human haemoglobin agglutinate when they react with faeces samples containing human haemoglobin.
Photometric measurement of the turbidimetric complex at 700 nm.
25 – 1000 µg/l (LOT dependent)
Samples with a higher concentration than 1:5 can be diluted with 0.9% NaCl and tested again.
It is recommended that each laboratory establishes its own cut-off according to the local population and the purpose of the assay (e.g. screening, follow-up, etc.).
Sensitivity and specificity vary depending on the cut-off as follows:
The test takes approximately 10 minutes.
For sample collection the iFOB sample collector kit is required.
Order No.: ST0205
Please note: All package inserts provided on this website are intended for information purposes only! Always process all tests, controls and calibrators strictly according to the package insert provided with the kit.
Find the evaluation study for the Eurolyser iFOB assay here: Evaluation Eurolyser iFOB
Please note: depending on the time of the market launch of an assay, the respective assay evaluation was carried out using either a Eurolyser smart instrument or a Eurolyser CUBE instrument.
Since the measurement method is the same in both instruments (their photometric units are identical in construction), as was proved in this comparison study, all testkit evaluations provided on this website are valid for either instrument!
ST 0200 iFOB (faecal occult blood) test kit (containing 32 tests)
ST 2000 iFOB (faecal occult blood) control kit (2 x 2 ml (high and low) )
ST 0205 iFOB (faecal occult blood) collector kit (32 collectors)